Objetivo: ISO specifies the properties and performance of sterile single-use hypodermic syringes with or without needle, made of plastic materials . Find the most up-to-date version of ISO at Engineering Sterile hypodermic syringes for single use – Part 3: Auto-disable syringes for fixed -dose immunization (ISO ); German version EN ISO
|Published (Last):||2 February 2006|
|PDF File Size:||11.12 Mb|
|ePub File Size:||15.71 Mb|
|Price:||Free* [*Free Regsitration Required]|
The details of conditions under which the licence for the use of the Standard Mark may be granted to manufacturers or jso may be obtained from the Bureau of Indian Standards.
МКС EN ISO 7886-3:2011
Attention is particularly drawn to the following: Re-use of injection equipment in the absence of sterilization has increasingly let to transmission of blood-borne pathogens this standard. Autodisable syringes for fixed-dose immunization ISO Maintain the water and the syringe 7886-3 this temperature. The corresponding Indian Standards which are 78886-3 be substituted in their places are listed below along with their degree of equivalence for the editions indicated: It does not address compatibility with injection fluids.
This part of ISO does not specify the design of the auto-disable feature, which is left to the discretion ofthe manufacturer. The preparation of this standard was recognized as a high priority requirement to prevent the re-use of fixed dose immunization syringes in the developing and transitional countries.
BS EN ISO 7886-3:2009
This standard is intended to cover ‘fixed dose’ immunization syringes that are rendered inoperable after delivery of the intended dose. It should not be possible to defeat the auto-disable feature by removing and re-inserting the plunger.
The piston shall not become detached from the plunger when tested in accordance with Annex B of ISO No secondary or additional action on the part of the user shall be required. Iiso the plunger travel and readjust the recorder to zero.
The projection of the plunger and the configuration of the push-button should be such as to allow the plunger to be operated without difficulty. Then click on “Details” and select the required version.
It is important to us that you purchase the right document. Water for analytical laboratory use — Specification and test methods ISO These lines shall be of uniform thickness. Needle tubing shall be in accordance with ISO Using the test device C.
Buy Do you want to buy this document? For kso with integrated needle, 14,2 of ISO This Indian Standard has been developed from Doc No.: International Standard ISO Increase the force to a maximum of N or until the plunger is capable of delivering a second dose from the same syringe.
NBN EN ISO | NBN
Syringes with re-use prevention feature This document has been re-assessed by the committee, and judged to still be up to date. Connect the nozzle of the syringe to the tubing of the reservoir B. This part of ISO does not specify the design of the auto-disable feature, which is left to the discretion of the manufacturer. It is recognized that syringes designed to reduce the risk of needlestick injuries, in addition to preventing sharps injuries, may also comply with this part of standard with regard to their auto-disable properties, but it is stressed that anti-needlestick properties of syringes are not in themselves addressed in this standard.
Search the history of over billion web pages on the Internet. You will be redirected to our e-shop where you can buy the document.
This part of JSO is not applicable to syringes for use with insulin specified in ISOsyringes made of glass specified in ISOsyringes for use with power-driven syringe pumps specified in ISOauto- disable syringes for variable dose delivery and syringes designed to be prefilled. For undated references, the latest edition of the referenced document including any amendments applies. They shall lie in planes at right angles to the axis of the barrel. In all cases, once the auto-disable feature has been activated: When a syringe with integrated needle is tested in accordance with Annex 8 of ISO Other standards can be applicable when syringes are used for any other intended purpose than those specified in ISO IS Part 3: Follow these links to find other Standards similar 7886-3 the one you are viewing.
Do you want to buy this document?