to reading the User’s Manual. Please read the Accutrend. Plus User’s Manual carefully before using the meter for the first time. The Accutrend Plus meter is used. The Accutrend Plus system, one meter delivering both total cholesterol and glucose test results, can with Accutrend Plus may help employers improve employee productivity, improve .. 15 Cholestech LDX System User Manual. Inverness. View and Download Cobas Accutrend Plus user manual online. Accutrend Plus Blood Glucose Meter pdf manual download.
|Country:||Bosnia & Herzegovina|
|Published (Last):||20 July 2013|
|PDF File Size:||3.50 Mb|
|ePub File Size:||6.65 Mb|
|Price:||Free* [*Free Regsitration Required]|
Close the container immediately after removing strip to protect the desiccant and strips, otherwise, the test strips may become unusable before their expiration date.
Performing a measurement Switching the code display You can switch between the display of the stored code numbers for information purposes. Preparing To Perform A Lpus Performing a measurement Preparing to perform a measurement Get the test strip container for the required measurement.
Quality control recommendation Applying the control solution 10 Apply a large free hanging drop of either the glucose or cholesterol low level control solution directly from the bottle to the manial strip. Quality control recommendation Starting the measurement 11 Close the measurement chamber flap. Memory Press the Set button to switch between the memory areas of the two test parameters.
COBAS ACCUTREND PLUS USER MANUAL Pdf Download.
Performing a measurement Performing a measurement After it is switched on and coded, the instrument expects you to insert the test strip.
Hold the test strip with thumb and index finger so that the printed measurement parameter is facing upwards. Press the M button until the desired month is displayed. Instrument settings The date and time as well as the set symbol now appear on the display.
Flap open displayed after instrument has been powered on The measurement chamber flap is not closed properly. About This Manual About this manual About this manual Page layout The page layout of this manual enables you to easily locate the most important information. Check that the whole display is functioning correctly to prevent misinterpretations due to defective display elements.
Instrument settings Setting the date With the next three settings, you first enter the year, then the month and finally the day. If you leave the cap off the container of test strips. Page Alphabetical index Time Instrument settings Setting the time First enter the current hour and then the minutes.
The drop of blood must be applied to the test strip immediately after sccutrend the fingertip. Page 9 Contents Quality control recommendation Preparing to perform a control test Page 19 About this manual Example of a display screen: Performing a measurement Flagging the measurement as a control: Explanations on flagging this measurement events, ev.
Flap open displayed during measurement The measurement chamber flap is not closed properly.
Check the expiration date of the test strip. Performing a measurement Assigning additional information to a measurement: You can flag a measured value at different points in time: Error messages Error messages In certain circumstances error messages may appear on your display.
Test strip is unusable The test strip has already been used or is dirty.
Page 66 Performing a measurement The flashing drop symbol above the strip symbol now instructs you to apply the blood. Performing a measurement Start measurement by closing the Display of measurement results measurement chamber flap Get the proper control solutions for the test strips. Performing a measurement Pkus notes Always … accuternd the instrument at the acceptable test-specific temperatures see also the test strip inserts: Alphabetical index Operating the instrument Quality control recommendation Display of results When measurement is finished, the result is displayed.
Cobas Accutrend Plus User Manual
Operating the instrument Instrument settings Brief overview of the instrument settings The following table provides an overview of the available settings. Every time the instrument is powered on, you can check the display.
If the entire memory is empty, three dashes — — — will appear in the display. The date and time accutend the display match the time at which the measurement was performed, and not the current time. Important Notes Performing a measurement Important notes Always … operate the instrument at the acceptable test-specific temperatures see also the test strip inserts: